Powered Air Purification Respirator
FROM WIN HEALTH LABS, LLC
A joint project to help provide much-needed respirators to the heroic men and women in healthcare who are working on the frontlines to combat the spread of COVID-19.
The respirator, created jointly by multiple companies, and manufactured and sold through WIN Health Labs, LLC is personal protective equipment (“PPE” ) with the ability to filter the air for the wearer. It is a powered, air-purifying respirator, or PAPR, that takes the place of a traditional medical mask and visor and features a replaceable polyethylene barrier that can be changed out by the user to reduce the amount of necessary decontamination.
GETTING STARTED WITH YOUR PAPR
Cautions and Limitations
– A –
– B –
– C –
– F –
– I* –
– J –
– L –
– M –
– N –
– O –
– P –
– S –
– PH –
Contains electrical components and the intrinsic safety has not been evaluated and approved by Mining Safety Health Administration (MSHA) or a recognized independent laboratory.
Instructions for Use
Check all equipment for damage prior to each use. Do not use damaged equipment. Ensure that all consumables are available prior to each use. Do not use equipment unless all consumables are available. Use the manufacturer’s supplied components only. For detailed information on instructions, please visit www.WINHealthLabs.com
- Remove Protective films (both sides) from the Face Shield Assembly
- Clean the Air Distribution Assembly as per the standard cleaning procedures prior to use.
- Fully charge the battery by plugging the wall adapter into a 120V receptacle and plugging the charger cord into the 12V In/Out port of the battery. Turn the battery pack ON (-) to charge. Battery is fully charged when all four indicator lights are lit.
- Remove the filter from the package. Open filter housing by releasing the two snaps on the housing and opening the cover. Remove shipping label/ sleeve on filter housing and insert filter into housing with flange side out.
NOTE: The recommended change interval for the HEPA filter is not to exceed 3 months, or if the filter shows signs of blockage or physical damage.
- Adjust straps according to individual fit preferences.
- Lock the Air Distribution Assembly on the shoulder harness plate
- Only use original manufacture parts and consumables.
- Verify full charge on the battery
- Install Battery in the battery compartment in the Air Distribution Assembly. Connect Power Cable to the battery.
- Turn on the battery switch (“-”) and verify airflow as per the procedure listed in the guidelines below
- Put on shoulder harness with the Air Distribution Assembly
- Put Face Shield Assembly on head and adjust to the head size of the user
- Put on Hood with corners above shoulders
- Place Drawstring over hood
- Cinch Hood, allowing room for two fingers
- Tape down corners of the Hood to rear while Hood is filled with air
- Remove and disinfect drawstring and drawstring clip.
- Untuck Hood from neck area
- Remove Hood from corners away from body and discard
- Remove Shoulder Harness with the Air Distribution Assembly
- Remove Face Shield Assembly
- Disinfect all parts as per procedures listed in the guidelines below
- Remove Battery from the Air Distribution assembly and prepare for recharging
Guidelines for cleaning and disinfecting the PAPR Assembly
Your employer should thoroughly review this information prior to selecting the disinfecting product for your equipment and specific application. This PAPR has not evaluated the effectiveness of any agents with regards to inactivating viruses on the PAPR or its components.
Components of the PAPR may wear damaged over time with prolonged or extended use of disinfecting products. If you discover any signs of damage, remove the component from service and discard and replace or repair as appropriate, following the guidance in the product User Instructions. Do not use the PAPR after the expiration date.
Cleaning and/or disinfecting the Powered Air Purifying Respirator (PAPR) Assembly:
Use the following guidelines for cleaning your PAPR assembly. Additional inspection, cleaning, and storage procedures for Whirlpool’s PAPR components may be found below.
- It is important to follow all steps.
- Always practice good hand hygiene.
- Nitrile/vinyl gloves and other personal protective equipment should be worn during cleaning.
- With any disinfecting agent, follow the User Instructions in regards to usability, application and contact time, and ensure all components are thoroughly wiped with fresh, warm water.
- Ensure all components are dry prior to reuse or storage.
- Cleaned components should be carefully inspected per User Instructions to identify any damage, excessive wear, or deterioration of components and replace them as necessary. Store away from contaminated areas when not in use, and in accordance with the User Instructions.
Fan/blower unit and battery pack
- Do not allow liquid to enter the air outlet port or the motor housing area.
- Do not use compressed air or a vacuum to clean the interior of the fan/blower.
- Use caution cleaning around the battery pack connector pins where the battery seats on the bottom of the motor/blower unit. Ensure this area and pins are thoroughly dry before next use or storage.
- Clean outer surfaces of the PAPR assembly and battery pack with a clean soft cloth lightly dampened with warm ~43°C (110°F) water containing a mild pH neutral (pH 6-8) detergent. Do not immerse the fan/blower or battery pack during cleaning.
- Wipe the outer surfaces of the fan/blower assembly and battery pack with a clean soft cloth lightly dampened with one of the noted EPA-registered disinfectant cleaners. Follow the user instructions for the selected disinfectant. Do not soak the fan/blower assembly or battery pack.
- Wipe all outer surfaces with a clean soft cloth dampened with clean warm ~43°C (110°F) water.
- If the fan or battery cannot be properly cleaned, stop using the PAPR.
Headgear Assembly, Breathing Tube/Hose, and Shoulder Harnesses
- Prior to cleaning, discard hood and disassemble PAPR per User Instructions.
- Clean all parts with a clean soft cloth dampened lightly with warm ~43°C (110°F) water containing a mild pH neutral (pH 6-8) detergent.
- Wipe external surfaces with a clean soft cloth lightly dampened in one of the noted EPA-registered disinfectant cleaners. Follow user instructions for selected disinfectant.
- Clean components thoroughly with clean, warm, lightly damp cloth ~43°C (110°F) after each use.
- Components should be hand-dried with a clean lint-free cloth or air-dried.
- Allow to completely air dry prior to reuse or storage. Air dry in an uncontaminated atmosphere, temperature not to exceed ~43°C (110°F). Alternately, dry by connecting to the fan/blower unit and use it to force air through the tube until dry.
- NEVER ATTEMPT TO CLEAN THE FILTER. Replace and properly dispose of the filter when dirty or damaged.